Philips Respironics SimplyGo Portable Oxygen Concentrator
The Philips Respironics SimplyGo is a versatile, compact, and lightweight portable oxygen concentrator designed to help maintain an active lifestyle for people requiring long-term oxygen therapy. With its ability to deliver both continuous flow and pulse-dose oxygen, the SimplyGo is a standout device that offers the best in flexibility and efficiency. Weighing just 10 pounds, it’s designed to be rugged and easy to carry, making it perfect for users on the go. Its easy-to-understand control panel and large LCD screen ensure simplicity in operation, allowing users to adjust settings with minimal effort. The SimplyGo’s long-lasting battery provides up to 3.5 hours of use on a pulse setting of 2, offering users freedom and peace of mind while they’re out and about.
SimplyGo Key Features
- Versatile oxygen delivery modes: Provides continuous flow up to 2 liters per minute and pulse-dose delivery up to a setting of 6.
- Compact and lightweight: Weighs only 10 pounds, allowing users to maintain an active lifestyle.
- Long-lasting battery: Provides up to 3.5 hours of battery life on pulse setting 2, providing freedom and peace of mind for users.
- Easy-to-understand control panel: Large, easy-to-read LCD screen and intuitive control panel for simple operation.
- Rugged design: Built for the rigors of real-world use, with a durable shell and high-impact resistance.
|Weight||10 lbs (4.5 kg)|
|Dimensions||29.2 cm (H) x 25.4 cm (W) x 15.2 cm (D)|
|Oxygen Flow Rate||Pulse dose settings 1-6, continuous flow 0.5-2 LPM|
|Oxygen Concentration||87% to 96% at all settings|
|Oxygen Delivery||Pulse dose or continuous flow|
|Power Source||AC power supply (100-240 V, 50-60 Hz)|
|DC power supply (12-19 V, 8 A)|
|Rechargeable lithium-ion battery|
|Noise Level||43 decibels (at pulse setting 2)|
|Alarms and Indicators||System technical fault, Wrong battery, Low battery, Depleted battery, High breath rate, Low oxygen purity, No breath, No flow|
|Operating Altitude||Up to 10,000 feet (3,048 meters) above sea level|
|LCD Display||Color display showing oxygen settings, battery life, alarms, and indicators|
|Battery Charging||Approximately 2 to 3 hours from 0% charge to 100%|
|Filters||Includes Internal Bacteria Filter|
|Flow Settings||Continuous Flow: up to 2 LPM|
|Pulse Dose: 1-6 settings, in 1 increments|
SimplyGo Battery Duration
|Oxygen Setting||Pulse Dose||Continuous Flow|
|1||3.7 hours||2.3 hours|
|2||3.5 hours||0.7 hours|
Philips Respironics SimplyGo User Guide Video
This video delivers a comprehensive introduction for patients on how to initiate their journey with SimplyGo.
Included in the Package
Philips Respironics SimplyGo
- SimplyGo Concentrator
- Carryng Case
- Mobile Cart
- Ion-Lithium Battery
- DC and AC Power Cords
- Accesory Case
- User Guide
The SimplyGo POC weighs approximately 10 pounds (4.5 kg).
The dimensions of the SimplyGo POC are 29.2 cm (H) x 25.4 cm (W) x 15.2 cm (D).
The SimplyGo POC offers both Pulse Dose and Continuous Flow options. The Pulse Dose settings range from 1 to 6, while the Continuous Flow ranges from 0.5 to 2 liters per minute (LPM).
The battery duration varies based on the oxygen setting. For Pulse Dose settings, it can range from 1.9 to 3.7 hours, depending on the setting. For Continuous Flow settings, it can range from 0.7 to 3.1 hours.rward and cost-effective. However, it’s important to refer to the user manual or contact customer support for guidance to ensure proper replacement.
Yes, the SimplyGo POC is FAA approved for use on airplanes. However, it’s always recommended to check with the airline beforehand and follow their specific guidelines and requirements.
It takes approximately 2 to 3 hours to charge the battery from 0% to 100%.
The sieve beds in the SimplyGo POC generally require replacement every 12-18 months, or as recommended by the manufacturer. This timeframe may vary depending on factors such as usage, environment, and manufacturer guidelines.
No, the replacement of sieve beds typically requires the expertise of a trained technician or authorized service provider. It is advisable to follow the manufacturer’s instructions or contact Philips Respironics or their authorized service centers for professional assistance in replacing the sieve beds.
There are no reviews yet.